I need a minimum of 3 volts from my daq to control. My next attempt was to set some weight values in the center to zero “weights2 some zero-weights ” — blue dots with zeros at center time. They are excellent for research, development, produc t ion, and incoming inspection applications. Embed this content in your HTML. My first attempt was to introduce a weighting that would prefer the outer data instead of the central ones that is “weight1 no zero weights ” — red V-shaped curve.
|Date Added:||4 August 2005|
|File Size:||27.22 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
To address these applications the offers high speed, programmable dynamic loading, sweep simulation and control capability never before achieved. Cyroma addition, each module can be operated independently or synchronized so that all modules start operating at the same time while running independent programs.
Chroma ATE chr63200a Electronic Load
Not sure how to do it. The right figure shows the programmable parameters which can be set within the loads: Difficulty in plotting xy graph for x, y inputs coming from two different while loops one under another!
I found an example to generate an AVI file from a photo. This capability is ideal for determining worst case voltage peaks see Figure 1. The user can perform online voltage measurements and adjustments or simulate short circuit tests using the simple keypad on the front panel. Low voltage operation is possible towards zero volts with corresponding reduced current levels see de-rating curves.
I have to read the signal from that instrument but I don’t really understand how. There are mainly two things I don’t understand:. The series load modules operate in constant voltage, current, resistance, power, or impedance to satisfy a wide range of test requirements.
Feature is excellent for determining the peak current which can be sustained by converters. I have always used the same Modbus address to configure the IO server and there was never a need to change it until now.
Chroma ATE 63206
lbview The maximum power for a single mainframe is 2kW when five load modules are paralleled. I can set pulse frequency, pulse lenght and number of pulses for kWh. The result is that I have to generate an AVI file for each photo an video and after I have to concatenate them This gave errors, all datas must have the same size I would like to program the outputs so that they are disactivated if a password has been inserted e.
My chfoma is to eliminate the sharp peak in the input data white. Dedicated remote load sensors and control circuits guarantee minimum waveform distortion during dynamic loading.
There is no error code, nor missing filename. I also attached the front panel screen shot to have a view how the xy hraph appears during the measurement. At the minimum voltage 0. As a cherry on the top of it. I choma self taught my way throughout this project through the getting started guide and a lot of searching around the forums. How can I do that? I need a minimum of 3 volts from my daq to control.
Run into this error when install my application on another pc win xp. In addition, series also provides voltage peak measurements during actual loading conditions. Ajout variable dans tableau.
IVI driver – Discussion Forums – National Instruments
Ive attached 636000 the main VI file and the test VI that contains the password prompt. Avoiding the need for using an oscilloscope to capture the voltage peak, saving time and costs. Add to Inquiry Cart Data Sheet.
The also has a dynamic sweep function to meet the test requirements of ATX power supplies. So, I deleted I set some values in the amplitudes to NaN — those values which are part of the sharp peak.